2 posts • Page 1 of 1
- Regular User
- Posts: 17
- Joined: Mon Oct 19, 2015 10:34 am
- Location: Waller, Texas (Outside Houston)
Wondering if anyone ever heard of this problem and know of a fix. On the Handbox, for the Joystick Touch Speed, when button is pressed and light is on, the probe should move very slowly, right? Well, when I try to use the feature this time, instead of creeping slowly, the Probe moves very fast with no braking. I have to physically grap it before limiting out. It had always work until now and I cannot fiqure it out. Anyone have any suggestions? Integra 10.10.8
You do not have the required permissions to view the files attached to this post.
I would reach out to LK Metrology... I assume a value in LKCMMDRV.CFG file has changed. There are parameters in this file that control Min, Max, Active, Default, etc. parameters related to motion, speed, acceleration, etc.